4-point rigid brace for treating injuries of the posterior cruciate ligament
M.4s PCL dynamic is a functional knee orthosis. The product allows the conservative or postoperative treatment of patients with a (partial) rupture of the posterior cruciate ligament.
By actively exerting a force via the PCL protection pad the tibia is holding in the desired position
Safe collateral stabilisation by rigid frame
Extension and flexion limitation protects against damaging joint movements and relieves injured structures
Optimal imitation of knee movement by using patented physioglide Hinge Technology
- Extension limitation: 0°, 10°, 20°, 30°, 45°
- Flexion limitation: 0°, 10°, 20°, 30°, 45°, 60°, 75°, 90°
- Immobilisation: 0°, 10°, 20°, 30°, 45°
Open all straps and pull them through the belt eyes. In order to prevent the the Velcro® from causing damage to clothing, for example, fold the end of the strap back on to it.
Pull the knob out of the tibia support. A click confirms the correct position. Pull the tibia support out towards the back until it stops. Bend your leg to ~70° and put the that you have put the brace on properly. The brace has been put on properly if the joint centres are located at the top edge of the kneecap and they are positioned slightly behind the side central line of the leg. Now fasten Belt No. 3 for the thigh and pull it tight. Make sure that you don’t position the brace too far forwards.
Turn the knob back in to activate the turning mechanism. Another click will confirm the correct position. Now turn the wheel to the extent that the tuberosity of the tibia (protruding bone at the top end of the shin) is level with the one on the healthy leg.
When required and for purely stabilising purposes, please fasten the tibia belt. This must always be the last belt to be fastened.
Now move the knee lightly a few times and check again that your have put the brace on properly. Make any necessary corrections.
The indicators on the side on the guide bars may be used as a reference point for the position of the tibia support.
Only change the extension and flexion limits on instruction from your doctor or orthotist. In order to prevent hyperextension, always insert extension wedges or 0° wedges. The wedges define the exact excursion of the orthosis. In difficult soft-tissue conditions, e.g. where the soft-tissues are under compression, it may be necessary to insert the larger wedge to limit the joint movement to the desired excursion. In addition, make sure that the orthosis joints are correctly positioned: the middles of the joints should be level with the upper border of the patella. The joint must lie behind the midline of the leg.
Avional, Polyamide, PU, Polyester
Soap residues, lotions and ointments can cause skin irritation and material wear. Wash the product by hand, preferably using medi clean washing agent.
- Do not bleach.
- Leave to dry naturally.
- Do not iron.
- Do not dry clean.
With normal use, the hinge mechanism should not require any maintenance, but it can be lubricated with Teflon spray. Check regularly that the joint screws are secure, tighten them if necessary.
- Conservative treatment after injury of the posterior cruciate ligament
- Postoperative treatment after reconstruction of the posterior cruciate ligament
M.4s® PCL dynamic instructions for use 4.70 MB
How to use our M.4s PCL dynamic knee braces. Date of issue: until December 2017.
medi knee-range 4.21 MB
More flexibility in treatment
M.4s PCL dynamic instructions for use 4.77 MB
How to use our M.4s PCL dynamic knee braces. Date of issue: from December 2017.
M.4s® PCL dynamic – The new standard for PCL therapy
M.4s PCL dynamic – Instruction video for patients
More information about our new M.4s PCL dynamic knee brace