Functional rigid brace with flexion / extension limitation
The M.4s comfort is a functional knee orthosis. The product is exclusively to be used for the orthotic fitting of the knee and only on intact skin.
Optimal stabilisation using 4-point principle
Safe collateral stabilisation by riged Frame
Extension and flexion limitation protects against damaging joint movements and relieves injured structures
Optimal immitation of Knee movement by using patented physioglide Hinge Technology
- Extension limitation: 5° Hypex, 0°, 10°, 20°, 30°, 45°
- Flexion limitation: 0°, 10°, 20°, 30°, 45°, 60°, 75°, 90°, 120°
- Immobilisation: 0°, 10°, 20°, 30°, 45°
Open all the straps. Sit on the edge of a chair, pull the brace up the affected leg and flex your
knee to 45°. Position the joints so that the centre of the joint (extension wedges’ height) is positioned at the level of the top edge of the kneecap and slightly behind the leg’s side central line. Please make sure that the brace has been placed on the leg straight and is not twisted.
First fasten the lower front calf strap, then the upper back thigh strap. Keep your knee in a flexed position and push both joints towards the back of the leg.
Now fasten the lower back thigh strap. Make sure that you do not have the hinge joints positioned in front of the midline of the leg.
Now fasten the upper front calf strap. Then the upper back calf strap, and finally the front thigh strap.
Now move the knee lightly a few times and check again that your have put the brace on properly. Make any necessary corrections
Only change the extension and flexion limits on instruction from your doctor or orthotist. In order to prevent hyperextension, always insert extension wedges or 0° wedges. The wedges define the exact excursion of the orthosis. In difficult soft-tissue conditions, e.g. where the soft-tissues are under compression, it may be necessary to insert the larger wedge to limit the joint movement to the desired excursion. In addition, make sure that the orthosis joints are correctly positioned: the middles of the joints should be level with the upper border of the patella. The joint must lie behind the midline of the leg.
Shaping the orthosis
When shaping the orthosis, please ensure that the joints stay parallel. This guarantees that the orthosis will function well and run easily and that there is no excessive wear.
Fastening the belt eyes
The belt eyes have been adequately fastened and are, to some extent, deliberately mobile. Please do not tighten the screws; they are tight enough. If you apply too much force, you can break off the screws.
Avional, velours, polyester, PU foam
The brace cannot be worn in saltwater. If the brace is worn in chlorinated water, it should be rinsed out thoroughly with clean water afterwards. Soap residues, lotions and ointments can cause skin irritation and material wear.
Wash the product by hand, preferably using medi clean washing agent.
- Do not bleach.
- Leave to dry naturally.
- Do not iron.
- Do not dry clean.
With normal use, the hinge mechanism should not require any maintenance, but it can be lubricated with Teflon spray. Check regularly that the joint screws are secure, tighten them if necessary.
- Nonsurgical / postoperative after cruciate ligament ruptures (isolated or with accompanying injuries)
- Nonsurgical / postoperative after collateral ligament ruptures
- Nonsurgical in the case of knee instability (isolated or combined)
- Postoperative after meniscus intervention
- Postoperative after cartilage intervention
M.4s® comfort instructions for use 3.23 MB
How to use our M.4s comfort 4-point knee braces. Date of issue: until December 2017.
medi knee-range 5.48 MB
More flexibility in treatment
M.4s comfort instructions for use 3.50 MB
How to use our M.4s comfort knee orthoses. Date of issue: until February 2018.
M.4s® comfort instructions for use 3.52 MB
How to use our M.4s® comfort knee braces. Date of issue: until May 2018.
M.4s® comfort instructions for use 3.12 MB
How to use our M.4s® comfort knee braces. Date of issue: from May 2018.