Walker with range of motion control and fixed walker
protect.ROM Walker is a lower leg brace. The product is exclusively to be used for the orthotic fitting of the foot and only on unbroken skin.
protect.ROM Walker with range of motion hinge (very easy to use)
Available in 4 sizes, adjustable circumference
Velcro system facilitates wound inspection and personal hygieneRange of motion hinge (protect.ROM Walker)
How does it work?/Mode of action
Optional fixed joint angle or partial excursion; this permits early functional mobilisation
Adjustable plantar flexion, dorsal flexion and immobilisation
Rocker sole encourages natural gait
Open all straps and remove the foam inner shoe from the Walker.
Open the leg and foot part of the inner shoe by loosening the Velcro fasteners. Ask the patient to place his foot and lower leg into the inner shoe.
Close the shoe and make sure that the foam fits closely against the patient's leg.
Make sure that the plastic sleeves over the splints still cover the Velcro strip on the internal sides of the splints in full. Ask the patient to place their foot in the plastic shell between the two stabilising bars. The foot should be positioned so that both of the splints are centred to the centre line on the side of the leg.
The splints can also be straightened for an optimum fitting and to ensure a close alignment on the sides of the calf. Comfort padding can also be fitted on the interior sides of the splints (at the level of the ankle bone). These can be moved or removed to adjust the medial or lateral width between the splint and the foot.
If you have made sure that the splints are correctly adjusted and aligned (centre line on the sides of the leg), you can remove the plastic sleeves from the splints. The internal foam shoe is now held in place by the Velcro strip which is situated on the internal side of the stabilisation splints.
Close both of the forefoot straps which hold the patient's foot in place in the inner shoe. The straps should be firmly in place but can also be adjusted individually by the opening and closing of the Velcro fastener.
Any unwanted extra material can also be cut away.
Now close the three straps which hold the lower leg in position between the splints.
Start with the strap which is nearest to the ankle. The three straps should fit tightly.
Ask the patient to walk around in the orthosis to make sure that the Walker is sitting properly.
Turn the opening of the plastic cover to the current position of the pins in order to fit the ROM- joint. Now you can set the desired degree for the dorsal extension and for the plantar flexion by simultaneuosly removing the pin and turning. The medial and lateral setup has to match.
Removing the boot
If the patient is allowed to remove the boot during their recovery, or when the boot is removed during a progress check-up, the straps should be released and the Velcro hook that closes the bootie should be opened to allow the patient to step out of the boot with the bootie still locked in position in the protect.Walker boot shell. This will enable the patient to easily reapply the bootie and boot with the proper leg alignment.
Polyester, Nylon, PU-Foam, Fibreglass, Polyamide, Aluminium
Velcro fasteners should be closed for washing. Soap residues, lotions and ointments can cause skin irritation and material wear.
Wash the paddings by hand, preferably using medi clean washing agent.
Do not bleach.
Leave to dry naturally.
Do not iron.
Do not dry clean.
- Conservative and post operative care of foot or toe injuries or surgery
- Grade 1, 2, or 3 ankle sprains
- Stable or internally fixed fractures of the foot or ankle
- Post ligament, soft tissue and tendon surgery
- For postoperative and conservative treatment of fractures of the fibula
- Dislocated fractures of the talocrural joint
- Conservative treatment of injuries of the fore and middle foot and of the talus
- After ligament refixation and peroneal tendon dislocation
- For chronic ligamentous instability
- Follow-up treatment following fractures of the talus, calcaneus and medial malleolus
- For early functional therapy after conservative or surgical treatment for rupture of the Achilles tendon
- +49 921 912-0
- +49 921 912-783
- Mon. to Tues. 08.00 - 17.00